I get an rrror when using USB-1208FS with XAIN.vi LabVIEW example program.
USB 13D02 The user has the wrong range selected in LabVIEW. When in SE mode, the 1208 modules only have a +/- 10V range, and the LabVIEW example program defaults to +/-5V. Error Messages UniversalLibraryforLabVIEWv6071UL All All en All
For the time being we are unable to offer the following product ranges although we are currently working hard to increase the number of products we can offer in the future. Please contact us to talk about alternative products that we may be able to offer you.