Results 1 - 8 - 8.
How does CE Wedge know where the end of the data record is?
CE-Wedge looks for a time delay between each data item to signal the end of the record. If you specify any Postamble keystrokes, these will not be sent until no more data has been received through the serial port for roughly a third of a second. CEWedge 1 en....
On starting CE-Wedge on my HP Journada 680 I get an error message stating that files are missing.
Problems installin CE-Wedge onto a HP Journada device. The HP Journada 680 is an older HP HandHeld PC (HPC). The most common version of CEWedge that Tal sell is for the Pocket PC which is a completely different to the Journada 680 Handheld PC. There is a HPC version of CEWedge available and you can....
Unable to AutoSync After Using CE-Wedge
CE-Wedge should not conflict with the ActiveSync software in any way unless you have CEWedge running on the Pocket PC for the serial port that you want to use to connect with the ActiveSync software. If you are using the serial cable to connect to ActiveSync, then you will need to either shut down....
Will it work with any Windows CE device?
The CE-Wedge program was developed using the HP Journada 548 (must be a Journada Pocket PC & not the Journada Handheld PC) and the Compaq iPaq 3630 however any Windows CE device (running version 3.0 or above) should work. All Windows CE computers come with at lease one built in serial port however....
Can the configuration for different devices be saved?
CE-Wedge always uses whatever parameters were set in it the last time that it was used. You cannot create different configuration files and save them to different files. The program is really just a stripped down version that only provides a subset of the features in our WinWedge program. It was p....
Using a Socket I/O card
I am trying to use CE-Wedge to collect data from a handheld chlorophyll meter. As this is a low power device I have installed a Socket I/O card onto my HP Journada and was wondering how to set up CE Wedge, which only requests which COM port, not the flashcard slot. All that you need to do is inser....
My HP Journada starts to slow up after I have been collecting data for over 1 hour.
After collecting data using CE-Wedge for a while (1-2hrs) the HP Journada with a Socket I/O card, starts to slow up. This is likely to be caused by the Socket I/O card device driver. If the Journada is powered down (either intentionally or by the power saving feature in CE) while the serial port is....
Can CE-Wedge be used to collect data from multiple instruments?
The answer to this depends greatly on what the box is that the devices are connected to and how it works. If it is a GagePort from GageTalker Corp. then you should be able to use CEWedge without too much trouble. If the device is a serial multiplexer, then there may be a way to use it with CEWedge....
Results 1 - 8 - 8.