Using multiple PMDs with A/D, D/A and Dig IO in Dasylab
USB AA6A Used in Dasylab IRQ mode: If you use A/D you can not use D/A or Dig. IO on one PMD You can use D/A and Dig IO at same time on one PMD You can use multiple PMDs and use A/D on one and then D/A and Dig IO on the other, the speeds are: A/D 1200Hz max sample rate (which is max continuous detailed) D/A 100Hz max update rate Dig IO - max frequency: could o/p a square wave with 18mS period, Used in Dasylab Polling mode: You are limited to sample rate
For the time being we are unable to offer the following product ranges although we are currently working hard to increase the number of products we can offer in the future. Please contact us to talk about alternative products that we may be able to offer you.