Using the Datashuttle Express analogue output -summary
Strawberry Tree 36C6 The description below refers to operation of the Datashuttle Express analogue outputs with Workbench and Dasylab software The software allows two modes of output operation Asynchronous and Synchronous. These are selected form the "Experiment Setup" in the software Asynchronous provides no timing reference, update thus variable. Experimenting with experiment sampling rate and turning off the "Real time Output" in the modules helps, by forcing the worksheet timing to the output but accurate waveform production is not possible. OK for simple DC level changes. Synchronous provides a good stable output, but only works with the "fill once" selected in the Experiment set-up. This fills the output with a block of samples, then outputs then and repeats, using that same section of samples, in a circular buffer arrangement - however the buffer cannot accept new data until the worksheet is restarted. The Synchronous "continuous fill" is not supported. StrawberryTree n/a en
For the time being we are unable to offer the following product ranges although we are currently working hard to increase the number of products we can offer in the future. Please contact us to talk about alternative products that we may be able to offer you.